Black moulded plastic insulators covering the Neutral banana jack terminals of this cable set, are brittle and may crack. The cracks may occur during the factory assembly process or through stresses expected from normal operation and reasonable use. The cracks can be very difficult to see under typical inspection methods and may extend from the banana jack shroud into the handheld area of the connector. Additionally, the cracks may spread and lead to further damage which could expose the metal parts of the connector and allow direct contact.
Crack may allow a voltage transient present on the mains conductor to arc to the user leading to electric shock from a hazardous live neutral conductor or a phase conductor causing serious injury or death.
What to do...
Please IMMEDIATELY STOP using your Recalled Cable. Please register your Recalled Cable following the below instructions and make it unusable by cutting it in half. You will be asked to confirm destruction of your Recalled Cable during registration. Your Recalled Cable will be replaced at no cost.
Visit the following webpage: https://fluke.com/en-au/support/safety-notices/17xxquadcable-recall.